- About us
- Ozone cleaning services
- Our works
- Frequent questions
- Book Now
RBI 105 ultraviolet bactericidal recirculator (with 2 15 W lamps) (hereinafter referred to as recirculator) is designed to disinfect the air in the presence of people. It is used in medical rooms, for children, sports, industrial, local and public institutions. It is an effective means of preventing and combating airborne infections (flu, tonsillitis, etc.).
2. TECHNICAL SPECIFICATIONS
The emission source is two 15W bactericidal lamps.
Average lamp burning time, hours - UV HNL G13 - 8000; OSRAM - 10800.
Supply voltage - 220V
Frequency, 50 Hz.
Electricity consumption - no more than 50 watts.
Protection class according to GOST 12.2.025-76 - class I type N.
Weight, kg - maximum 3,200.
Average length of service, years - 5.
Warranty period 12 months
3. DELIVERY SET
Recirculator with lamps, pcs. - 1
Remote control, pcs. - 1
Instructions for use, pcs. - 1
Lamp manufacturer certificate, pcs. - 1
The recirculator consists of a stainless steel housing, 2 lamps, electronic ballast, fan for recirculation air. The housing is made of stainless metal. The fan installed inside the housing draws air through the inner chamber of the recirculator, where it is cleaned of the bactericidal lamp, the recirculator has 2 lamps low-pressure bactericides, emits an ultraviolet spectrum with a wavelength of 253.7 nm, which is close to the maximum bactericidal effect of the radiant energy. This suppresses the vital activity of microorganisms in the air, vegetative viruses and bacteria (bacilli, cocci) sensitive to the effects of ultraviolet rays.
5. SAFETY AND ENVIRONMENTAL MEASURES CAREFUL!
Only personnel who have completed the necessary training are allowed to operate the air recirculator. A disconnect from the mains in the following cases: replacement of the lamp, ballast, removal of defects, disinfection and cleaning of dust.
If the lamp breaks, it is necessary to collect the drops of mercury with a pear rubber and wash the place where the lamp broke, with 1% solution of potassium permanganate. Throwing the lamps defects must be carried out in accordance with the “Methodological recommendations for monitoring the organization current and final demercurization and evaluation of its efficiency ”no. 545-87 of 12.31.87.
6. PREPARATION FOR WORK
After unpacking the recirculator, check its integrity. Install the recirculator in the room at a height at least 2 m from the floor in a horizontal position Insert the plug into a grounded socket. If the recirculator comes with a power cord, to connect it plug into a socket connected to earth. Press the ON button on the remote control. Nominal lamp to increase the time of initial exposure by 1.2 times (20%), and after 2/3 of the nominal duration of the service term - by 1.3 times. Exposure controls should be performed every 6-12 months (depending on the intensity of use) with a dosimeter type DAU-81 TU 46-13-1443-83 or with an SRP-86 band spectro-radiometer. Exterior finish of the recirculator, in addition to processing the lamp, allows wet sanitization.
7. CURRENT REPAIR
Routine maintenance must be performed by specialists of repair companies. During the repair, you must the safety measures specified in this instruction are observed.
8. MANUFACTURER'S WARRANTIES
The manufacturer shall ensure that the recirculator meets the technical requirements, if the consumer comply with the conditions of transport, storage and operation. The warranty period of operation is 12 months from the date of sale, but not more than 18 months from the date of manufacture. The warranty does not apply to the lamp. During the warranty period, the manufacturer repairs the recirculator and its parts upon presentation of the Instruction exploitation at the address: MD4101, Cimișlia, str. Ion Popușoi 16, R.Moldova.
9. CERTIFICATE OF ACCEPTANCE OF PRODUCTS
The ultraviolet bactericidal recirculator complies with the technical specifications of TU U 33.1-30108964-003: 2007 and s recognized as suitable for use by the National Agency of Public Health of the Republic of Moldova: NP- 7599/2020 of 21.10.2020, FOR USE IN THE PRESENCE OF PERSONS.